<!--
 * @Author: cgl
 * @Date: 2023-04-18 15:12:26
 * @LastEditors: cgl
 * @LastEditTime: 2023-09-04 15:23:28
 * @FilePath: \public-health-emergencies-fed:\新放射\新radiation-hygiene-fe\src\views\person\itemControl\item12.vue
 * @Description:
 *
 * Copyright (c) 2023 by cgl/微盛信息产业有限公司, All Rights Reserved.
-->
<template>
	<div>
    <table border="1" cellspacing="0" cellpadding="0" align="center" style="width: 100%;">
      <tr class="title">
        <td rowspan="2">序号</td>
        <td rowspan="2">检测项目</td>
        <td colspan="2">验收检测</td>
        <td colspan="2">状态检测</td>
        <td colspan="3">稳定性检测</td>
      </tr>
      <tr class="title">
        <td>检测条件</td>
        <td>要求</td>
        <td>检测条件</td>
        <td>要求</td>
        <td>检测条件</td>
        <td>要求</td>
        <td>周期</td>
      </tr>
      <tr>
        <td>1</td>
        <td>等中心偏差</td>
        <td>胶片法</td>
        <td>±1.0 mm</td>
        <td>胶片法</td>
        <td>±1.0 mm</td>
        <td>胶片法</td>
        <td>±1.0 mm</td>
        <td>六个月</td>
      </tr>
      <tr>
        <td>2</td>
        <td>治疗定位偏<br/>差</td>
        <td>模体靶点<br/>法</td>
        <td>≤2.0 mm</td>
        <td>模体靶点<br/>法</td>
        <td>≤2.0 mm</td>
        <td>模体靶点<br/>法</td>
        <td>≤2.0 mm</td>
        <td>六个月</td>
      </tr>
      <tr>
        <td>3</td>
        <td>照射野尺寸<br/>与标称值最<br/>大偏差</td>
        <td>胶片法，各<br/>准直器</td>
        <td>±1.0 mm</td>
        <td>胶片法，1<br/>档常用准<br/>直器</td>
        <td>±1.0 mm</td>
        <td>胶片法，各<br/>准直器</td>
        <td>±1.0 mm</td>
        <td>六个月</td>
      </tr>
      <tr>
        <td>4</td>
        <td>焦平面上照<br/>射野半影宽<br/>度</td>
        <td>胶片法，各<br/>准直器</td>
        <td>照射野直径<br/>≤20 mm 时，<br/>≤4 mm；<br/>照射野直径<br/>>20 mm 时，<br/>≤5 mm</td>
        <td>胶片法，1<br/>档常用准<br/>直器</td>
        <td>照射野直径<br/>≤20 mm 时，<br/>≤4 mm；<br/>照射野直径<br/>>20 mm 时，<br/>≤5 mm</td>
        <td>胶片法，各<br/>准直器</td>
        <td>照射野直径<br/>≤20 mm 时，<br/>≤4 mm；<br/>照射野直径<br/>>20 mm 时，<br/>≤5 mm</td>
        <td>六个月</td>
      </tr>
      <tr>
        <td>5</td>
        <td>等中心处计<br/>划剂量与实<br/>测剂量相对<br/>偏差</td>
        <td>模体法，各<br/>准直器</td>
        <td>±5 ％</td>
        <td>模体法，1<br/>档常用准<br/>直器</td>
        <td>±5 ％</td>
        <td>模体法，各<br/>准直器</td>
        <td>±5 ％</td>
        <td>六个月</td>
      </tr>
    </table>
	</div>
</template>

<script>
export default {
  name: "item55",
  data() {
    return {
      activeName: 'first'
    };
  },
  created() {
  },
  methods: {
  }
};
</script>

<style lang="scss" scoped>
@import './content.scss';
.longTable{
  .tableBody {
    display:block;
    height:50vh;
    overflow-y:scroll;
    scrollbar-width: none;
  }

  tbody,thead {
    display:table;
    width:100%;
    table-layout:fixed;
  }
  tbody, td{
    border-collapse: collapse;
    border: 1px solid #606266;
  }
}

</style>
